#### Navigation

Skip menu## Teslameter-fluxmeter universal **TPU-2V**.

**Purpose:**

- Teslameter-fluxmeter universal
**TPU-2V**is designed for measurement of magnetic induction of constant magnetic fields, maximal values of magnetic induction of alternating and impulse magnetic fields, flux linkage using connected to TPU-2V measurement coil. If a measurement coil with known turns count and mean turn square of cross section of coil is connected to TPU-2V, the instrument is able to measure magnetic flux through that coil and magnetic induction. If a Helmholz coil with known coil constant is connected to TPU-2V the instrument allows to carry out measurements of magnetic moment and obtain magnetization of sample that is placed into Helmholz coil - Teslameter-fluxmeter
**TPU-2V**can be used in precise analysis of magnetic fields, generated by magnetic systems and electrical devices of different purposes. Also it can be used in measurements of magnetic parameters of materials in closed or open circuit

**Main characteristics of TPU-2V in magnetic induction measurements mode with measurement zonds ****«M» и «S»** (mode **«Teslameter»**)

**«M» и «S»**(mode

**«Teslameter»**)

- Teslameter-fluxmeter allows for measurements of magnetic induction of constant magneti fields, maximal value of magnetic induction of impulse and alternating magnetic fields.
- Magnetic induction measurement range:
**from 0,01 to 2000 mT**on measrements range**1; 10; 100 mT; 1; 10; 100 T** - Magnetic induction indication range:
**from 1 mkT to 100 T** - Normal range of frequencies of magnetic induction of alternating magnetic field:
**from 5 to 2000 Hz**. - Duration of impulses of impulse magnetic field:
**from 100 mks to 2 s**, pulse rise time is**from 10 mks to 1 s**. - Limits of relative error of measurements δ, %, don't exceed values calculated by formulas listed below (Вr – current measurement range, мТл; Вm – measured value of magnetic induction, mT):
- magnetic induction value from
**0,01 to 20 mT**of constant magnetic field, measurement zonds**«M», «S»****δ = ± [0,5 + 0,1•(Вr/Вm - 1)]** - magnetic induction value more than
**20 to 2000 mT**of constant magnetic field, measurement zond**«M»****δ = ± [0,5 + 0,05•(Вr/Вm - 1)]** - magnetic induction value more than
**20 to 400 mT**of constant magnetic field, measurement zond**«S»**,**δ = ± [0,5 + 0,05•(Вr/Вm - 1)]** - magnetic induction value more than
**400 to 2000 mT**of constant magnetic field, measurement zond**«S»****δ = ± 0,5•(Вm/400)** - for measurements of maximal values of magnetic induction of alternating and impulse magnetic fields (base relative error δ0, %,)
**δ0 = ± [1,0 + 0,1•(Вr/Вm - 1)]** - Working range of frequencies of magnetic induction of alternating magnetic field:
**from 2 to 5 kHz**. - Limits of additional relative error of measurements of magnetic induction in working frequency range, δadd, %, don't exceed values calculated by formulas (f – dimensionless value, numerically equal to the frequency of frequency of magnetic induction being measured, kHz):
**δadd = ± [1,5•(f - 2)]** - Frequency range of alternating magnetic field magnetic induction indication:
**from 5 to 35 kHz**.

**Main characteristics in magnetic induction measuring mode using measurement zond ****«I»** (mode **«Teslameter»**)

**«I»**(mode

**«Teslameter»**)

- Maximal value of alternating magnetic field magnetic induction measurement range normalized to 50 Hz frequency
**from 10 mkT to 1 T**for measurement range**1; 10; 100 mT; 1; 10; 100 T**for frequencies in range**from 10 to 10000 Hz**. - Limits of relative error of measurement of magnetic induction of alternating magnetic field δ0, %, don't exceed values calculated by formula
**δ0 = ± [1,0 + 0,1•(Вr/Вm - 1)]**

**Main characteristics in measurement mode with measurement coils (mode ****«Fluxmeter»**)

**«Fluxmeter»**)

- Magnetic flux measurement range
**from 1 mkV∙s to 10 V∙s**form measurement ranges**0,1; 1; 10; 100 mV∙s; 1; 10 V∙s**in range of integrated voltages**from 1 mkV to 8 V**. - Limits of relative error of measurement of magnetic flux δ, %, don't exceed values calculated by formula (where Ψr – used measurement range; Ψm – measurement result):
**δ = ± [1,0 + 0,1 (Ψr/ Ψm - 1)]** - Диапазон показаний при измерениях потокосцепления
**от 0,1 до 1 мкВ∙с**. - Integration time
**100; 500 mks; 1; 5; 10; 50; 100; 200; 500 ms; 1; 2; 5; 10; 20; 30; 50 s**.

**Other capabilities of teslameter-fluxmeter TPU-2V**

- Teslameter-fluxmeter is capable of conduction series of independed measurements with measurement count from 1 to 10, calculation and printing output of result of every independent measurement in series, mean values and mean square error of mean value (MSE).
- Time of averaging is defined by discretization frequency taking into account internal prescaler. Time of averaging is chosen by operator and depending on type of measured magnetic field is set up by following parameters:
- constant magnetic field – time of measurement
**0,1, 0,8, 1,6, 6,4, 12,8, 25,6 s;** - alternating magnetic field – frequency range
**70, 35, 18, 8, 4, 2 kHz;** - impulse magnetic field – minimal duration of pulse
**6, 50, 100, 400, 800, 1600 mks**.

- constant magnetic field – time of measurement
- Teslameter-fluxmeter can be controlled by PC through USB by "Flux-U" soft from delivery set.